4-47. DISTILLATION TEST APPARATUS.
This task consists of:
INIT IAL SET-UP
General Mechanics Tool Kit, (Appendix B, Section III, Item 1)
Portable Electric Drill (Appendix B, Section III, Item 2) )
Twist Drill Set, (Appendix B, Section III, It em 2)
General Safety Instructions
Before performing any maintenance actions on electrical equipment, ensure all
electrical power has been turned off. Failure to comply with this warning may
result in serious injury or death.
Remove Shield Assembly. See Figure 4-43.
Unplug condenser (1) from wall receptacle.
Remove ten screws (2) securing shield assembly (3) to mounting bracket (4).
Lift shield assembly (3) up and forward disconnecting shield assembly from condenser assembly (1).
2. Remove Condenser Assembly.
Remove shield assembly as in paragraph 1 above.
Squeeze clamp (5) and remove drain line (6).
Remove two screws (7) securing connecting bracket (8) to both condenser assemblies (1).
Remove six screws (9) securing condenser assembly (1) to mounting bracket (10) and remove condenser.
Install Condenser Assembly. See Figure 4-43.
If necessary, using mounting plate (10) as template, scribe and drill holes in condenser (1).
Install mounting bracket (10) on condenser (1) and secure with six screws (9).
Install connecting bracket (8) and secure with two screws (7).
Squeeze clamp (5) and connect drain line (6).
Plug in condenser assembly and verify operation.