4-48. REMOVE/INSTALL OXIDATION STABILITY BATH.
This task covers:
General Mechanics Tool Kit, (Appendix B, Section III, Item 2)
General Safety Instructions
Before performing any maintenance actions on electrical equipment, ensure all electrical power
has (been turned off. Death or serious injury may occur from failure to do this.
Remove the Oxidation Stability Bath. See figure 4-48.
At Main Power Panel, place circuit breaker AlCB15 to the OFF position and unplug oxidation stability bath power
cord from wall receptacle located inside cabinet "SI".
Disconnect drain line on oxidation stability bath located inside cabinet "S1".
Remove one screw from bath junction box and remove cover.
Disconnect heating element leads.
Disconnect conduit and withdraw with leads from junction box.
Remove heater element with junction box from bath.
Remove eight screws, nuts, lockwashers, and flat washers that secure oxidation stability bath to countertop.
Remove oxidation stability bath.
Install Oxidation Stability Bath. See figure 4-48.
Place oxidation stability bath in opening of countertop.