4-46. REMOVE/INSTALL THE CENTRIFUGE-continued.
Install Centrifuge. See figure 4-46.
Place new centrifuge into cut-out on countertop.
Align shock mounts with screw holes on shelf and attach each shock mount with four screws, lockwashers, flat
washers, and nuts. Tighten in place.
Plug power cord into convenience outlet and at main power panel, place circuit breaker A1CB8 to the ON
Run operational test to check new centrifuge.
4-47. REMOVE/INSTALL THE DISTILLATION TEST APPARATUS.
This task covers:
General Mechanics Tool Kit, (Appendix B, Section III, Item 2)
Drill, (Appendix B, Section III, Item 1 )
Drill Set, (Appendix B, Section III, Item 1)
General Safety Instructions
Before performing any maintenance actions on electrical equipment, ensure all electrical power
has been turned off. Death or serious injury may occur from failure to do this.
Remove Shield Assembly. See figure 4-47.
Unplug the condenser from the wall receptacles.
Remove ten shield assembly mounting screws.
Lift unit forward and disconnect shield assembly from the condenser assembly.
Remove defective shield assembly.