4-44. REMOVE/INSTALL KINEMATIC VISCOSITY BATHS - continued.
Install the two hold down brackets with captive wing bolts.
Place bracket with strap in position on viscosity bath. Secure strap.
Plug bath into controller and plug controller for viscosity bath into wall receptacle.
Verify viscosity bath operation.
For component removal refer to TM 10-6640-226-13&P.
4-46. REMOVE/INSTALL THE CENTRIFUGE.
This task covers:
General Mechanics Tool Kit, (Appendix B, Section III, Item 2)
Lockwashers, (Appendix F)
General Safety Instructions
Before performing any maintenance actions on electrical equipment, ensure all electrical power
has been turned off. Death or serious injury may occur from failure to do this.
Remove the Centrifuge. See figure 4-46.
At Main Power Panel place circuit breaker A1CB8 to the OFF position.
Unplug centrifuge power cord from convenience outlet.
From beneath countertop, remove four mounting screws, lockwashers, flat washers and nuts that attach each
shock mount to shelf.
Remove defective centrifuge from countertop.
For component removal refer to TM 10-6640-230-13&P.